The background module has an extension interface. It can be used to add programmaticly images as background including live images, control the current view of the LayoutEditor and synchronize the aligment its with any other device. The interface to open source. Please contact sales to get details on the specifications including an example code and getting help on implementing an interface to your device. A commercial plug-in development can be offered.
An plug-in to access a SEM from TESCAN was introduced with release 20190110. To start the TECAN plug-in it had to be loaded in the same way as a normal background image (mainemenu/Utiltites/Misc/Set background image). This plug-in is shiped within the LayoutEditor package and located in the ‚LayoutEditor/bin‘ folder on Windows systems and in the ‚plugins‘ folder on Linux and Mac systems. The plug-in can also be load by a macro. As any other macro this macro can ba added anywhere in the menu tree of the LayoutEditor. A ready to use macro for this task is located under mainmenu/Utilities/Macro/Setup/Connect to TESCAN-SEM. Also it is possible to set this macro as an autostart macro triggered any time the LayoutEditor is started (in the setup dialog under macros/autostart macro)
Once the plug-in is loaded the background tools of the LayoutEditor will be display. Next to all other features the LayoutEditor offers to handel background images and new tab with controls for the TESCAN-SEM is display.
Host IP address and port number of the SEM can be entered and the connect button will established the connection to the SEM. On old Linux systems (e.g. for Ubuntu 14.4 Linux or RHEL6) the SEM interface had to be in the firewall friendly mode. On Windows,Mac and newer Linux system both connection modes of the SEM are fine.
Once the connection is established the tab interface switched to the SEM control:
With the display section the SEM stage position display within the layout can be controled.
After the first connection to the SEM the stage position is not aligned. There will be a offset between the design and the stage. However the scale will be correct. To perform the wafe/stage alignment the wafer align controls are used. An alignment can be made on your choice by a 1 point, 2 points or 3 points alignment:
The alignment of the single three points is identical. In the normal operation the SEM stage and the display of the SEM position in the design will move synchron. Once an alignment is started SEM stage and background image can be moved separated. The center of stage and the center of the background image had to be moved to the same location. The alignment of this point is completed once the align button is pressed.
An alignment error will be show up in case the resulting scale diverge from the SEM calibration more than 20%. The alignment for a design will be stored/restored for each design name. clear alignment will remove the stored values.
move stage to view will move the SEM stage to the position that is currently show in the layout window.
view stage position will move the current layout display to the current stage position. This is independent whether the stage position is displayed or not.
clone live image will create an additional background image with the current liven image. The display can be altered with any option normal background images have. In this way multi locations can be display behind the layout. For the cloned image as well as for the live image all display option for normal static background images are available.
Any settings are store on software shutdown and loaded on restart.